This document applies to the basic safety and essential performance of an EMS ventilator in combination with its accessories, hereafter also referred to as ME equipment:
- intended for patients who need differing levels of support from artificial ventilation including ventilator-dependent patients;
- intended to be operated by a healthcare professional operator;
- intended for use in the EMS environment; and
- intended for invasive or non-invasive ventilation.
NOTE 1 An EMS ventilator can also be used for transport within a professional healthcare facility.
* An EMS ventilator is not considered to utilize a physiologic closed loop-control system unless it uses a physiological patient variable to adjust the ventilation therapy settings.
This document is also applicable to those accessories intended by their manufacturer to be connected to the ventilator breathing system, or to an EMS ventilator, where the characteristics of those accessories can affect the basic safety or essential performance of the EMS ventilator.
NOTE 2 If a clause or subclause is specifically intended to be applicable to ME equipment only, or to ME systems only, the title and content of that clause or subclause will say so. If that is not the case, the clause or subclause applies both to ME equipment and to ME systems, as relevant.
Hazards inherent in the intended physiological function of ME equipment or ME systems within the scope of this document are not covered by specific requirements in this document except in IEC 60601-1:2005, 7.2.13 and 8.4.1.
NOTE 3 Additional information can be found in IEC 60601-1:2005+AMD1:2012, 4.2.
This document does not specify the requirements for the following:
- ventilators or accessories intended for ventilator-dependent patients in critical care applications, which are given in ISO 80601-2-12.
- ventilators or accessories intended for ventilator-dependent patients in the home healthcare environment, which are given in ISO 80601-2-72.
- ventilators or accessories intended for anaesthetic applications, which are given in ISO 80601‐2‐13.
- ventilators or accessories intended for ventilatory support equipment (intended only to augment the ventilation of spontaneously breathing patients), which are given in ISO 80601‑2-79 and ISO 80601-2-80 .
- obstructive sleep apnoea therapy ME equipment, which are given in ISO 80601‐2‐70.
- operator-powered resuscitators, which are given in ISO 10651‐4.
- gas-powered emergency resuscitators, which are given in ISO 10651‐5.
- continuous positive airway pressure (CPAP) ME equipment .
- high‐frequency jet ventilators (HFJVs), which are given in ISO 80601-2-87.
- high‐frequency oscillatory ventilators (HFOVs), which are given in ISO 80601-2-87.
NOTE 4 An EMS ventilator can incorporate high-frequency jet or high-frequency oscillatory ventilation-modes.
- cuirass or "iron‐lung" ventilators.
 ISO 80601-2-79 and ISO 80601-2-80 replace ISO 10651-6, which has been withdrawn.
Status: PublishedPublication date: 2020-07
Edition: 1Number of pages: 110
Technical Committee: ISO/TC 121/SC 3 Respiratory devices and related equipment used for patient care
- ICS :
- 11.040.10 Anaesthetic, respiratory and reanimation equipment
This standard contributes to the following Sustainable Development Goal:
Buy this standard
|std 1 208|
|std 2 208||Paper|
A standard is reviewed every 5 years
Stage: 90.92 (To be revised)
Will be replaced byUnder development
Got a question?
Check out our FAQs
Monday to Friday - 09:00-12:00, 14:00-17:00 (UTC+1)
Keep up to date with ISO
Sign up to our newsletter for the latest news, views and product information.