This document specifies the requirements and test methods for mobile dental units concerning:
a) the properties of mobile dental unit connections to the compressed air supply, water supply, suction supply, and wastewater drain plumbing,
b) the materials, design, and construction of the compressed air and water system within the mobile dental unit,
c) the quality for incoming water and air,
d) the performance of mobile dental unit suction system, and
e) the air, water, suction and wastewater properties of mobile dental unit connections to the interfaces to dental handpieces.
This document also specifies requirements for instructions for use and a technical description.
This document is only applicable to mobile dental units that are not used for oral surgery treatment requiring sterile air and water supplies. Amalgam separators are not included in this document.
Status: PublishedPublication date: 2022-09
Edition: 1Number of pages: 13
Technical Committee: ISO/TC 106/SC 6 Dental equipment
- ICS :
- 11.060.20 Dental equipment
This standard contributes to the following Sustainable Development Goal:
Buy this standard
|std 1 92||PDF + ePub|
|std 2 92||Paper|
ISO 5467-2:2022Stage: 60.60
Got a question?
Check out our FAQs
Monday to Friday - 09:00-12:00, 14:00-17:00 (UTC+1)
Keep up to date with ISO
Sign up to our newsletter for the latest news, views and product information.